With the work on the Micro-Tel SG-811 generator PLL mostly completed, some trials with the Micro-Tel 1295 receiver – this instrument has similar YIGs fitted, just needs to be tuned 30 MHz above the actual frequency tuned, because the 1295 is running on a 30 MHz IF (all diagrams have tuned frequencies, not LO frequencies).
After some crude analysis of the schematics, the 1295 seems to be able to handle a bit more PLL bandwidth – so the target set more in the 500 Hz to 1 kHz region, and some calculations were carried out with the ADIsimPLL program, to determine the rough capacitor and resistor values – otherwise, the loop filter is the same as for the SG-811 PLL, also using an OPA284 opamp.
Otherwise, pretty much comparable results (earlier post related to the SG-811), for example (17.8141 GHz tuned/17.8171 GHz LO frequency, Icp setting 6):
After quite a few of these measurement (doesn’t actually take too long), the results.
adf41020 pll bw phase margin 1295
Again, a bandwidth frequency^0.7 product could be used to get the numbers down to two parameters – slope and intercept of the bandwidth*frequency^0.7 vs. Icp setting curve.
Finally, suitable Icp settings for a 600 Hz target BW:
The result seems quite satisfactory, pretty much constant 600 Hz BW can be achieved over the full 2 to 18 Ghz range, at about 47 degrees phase margin. This should allow for stable operation. No locking issues were observed at any of the frequencies, even with full Icp current.