A quick look at a really nice piece of kit, a 100 V, precision regulated power supply, can be floated to +-240 V, and can provide 100 Watts of power, or sink power, about the same range.
The front panel and handling is like any other HP system power supply from this era, and there are models 6632A (20 Volts) and 6633A (50 Volts) that share virtually the same control circuit. All is GPIB controlled, of course, and this unit has front and read outputs. I am going to use it for a capacitor tester (to study the voltage bias effect and hysteresis of ceramic capacitor materials), so I need a fairly reliable unit because it will run unattended for a while.
The top view, there is a massive heatsink, for 100 Watts of dissipation…
The transformer, it is the highest standard and insulation I have ever seen.
There are 8 power transistor, in a really massive output stage (4 complementary pairs, 2 each on each heatsink-the heatsink is sub-divided in two sections), each of them capable of handling 250 Watts of dissipation.
The output stage, it is a really generous design, considering that this is a 1 Amp supply (most designers would handle it with two transistors).
The only thing I don’t like about the unit, the RIFA X and Y rated caps. These are all cracked (still not shorted, but I don’t want to take a chance). So these will need to be replaced.
Otherwise, all is good with this unit, almost no dust inside – I believe this instrument had very low hours, or has been used in a very clean environment (not even a trace of dust on the fan).