Thanks to a kind contributor there is a new way of fixing the HP step attenuators that are ubiquitous in the various HP and Agilent generators, analyzers.
These attenuators exhibit some common failures modes-
(1) blown pads – fix by replacing with pads from good donor units. Keep in mind that even if the pad value is the same, there are pads of different geometry/length!
(2) mechanical issues with aged O-rings, easy to fix, best use some FKM O-rings
(3) the broken-off contact fingers, difficult to fix unless you have some precision equipment like a good milling machine or fine drill.
Here is an alternative way to fix it – use some two component epoxy glue after removing the remaining plastic parts from the contact finger by heat (heat gun or hot plate, about 200 degC).
To ensure a permanent repair, a piece of FR4 is affixed with epoxy glue. Sure enough this is not quite ideal for high GHz frequencies, but no problem up to 2 or 3 GHz.