This is another one of the marvelous HP 8662A generators, I got it eventually for free because it had parts missing but otherwise, a good unit.
The missing parts were the mains filter, and the transformer for the OCXO. The filter, I just replaced it with a fairly standard commercial unit, should be good enough for now. The transformer it is a bit more tricky, because of the limited space and rate type. By luck and coincidence I found a spare in Japan. 25 Euros, fair enough.
Be aware that there are many versions of the input assemblies – and my unit has a 1982 vintage power supply, with a 1996 back plane. This is not going to work well, so I needed to study all the wires and connections. No wonder someone had given up on the repair job before.
The later version:
Mounted these parts – yet, I didn’t turn the unit on, because of a blown fuse. After some inspection of the power supply, the common issues. A few bad caps on the driver assembly, replaced with a pair of new 22 uF axial caps.
Other issues – one of the main switching transistors is dead – ordered some spare transistors, new old stock, from Greece, and took them to Japan during the last business trip to Germany… these are MJ16012 power transistors. 800 Volt, 15 Amp 175 Watts. I don’t trust any copies of these, but rather bought good old Motorola parts. 1986 vintage.
Good practice with the 8662A power supply repairs – checked the control assembly.
The oscillator is working, 40 kHz present, but no drive output – how come? Some quick checks clearly show issues around the U1 inverter, a defective logic gate. Someone must have tried to fix it before, but seems he didn’t have the proper tools and solder – all sticky and bad, so I cleaned it with alcohol.
Desoldered the U1 4049 CMOS, with quite some difficulty because the drill used was quite small, and difficult to desolder.
Some strange observation – all the versions and date codes. The power supply control assembly, maybe 1997 vintage. The power supply base assembly and low voltage circuit, 1982 vintage. The A7A3 power converter assembly, 1993 with some parts dating back to 1986…
The serial – 34xx would suggest 1960+34=1994, but this can’t be, because the frame was only made past 12/1995.
Checked two random assemblies – also these were made in 1997.
Probably fair to say, a 1997 unit, and someone tried to fix it some time after that with a 1982 supply. Why he changed the power supply baseboard, and the removed the oven transformer, no clue.
The defective gate of the control assembly – order a few new CMOS circuits. Plenty of them back home in Germany, but here, it is faster and cheaper to order them new from Thailand… in addition, ordered a few 4013 and so on just in case.
Lastly, this unit also has an attenuator missing. Not uncommon for the 8662A. It is a 33321-60028 65 dB 5 V attenuator. 5/40/20 dB elements. Maybe I will find one on eBay, but actually, I have not much use for it because this 8662A will be running at +10 dBm all the time as a reference LO for my VCO test rig and phase noise analyzer.
One thought on “HP 8662A Synthesized Signal Generator: An almost new generator, with a pretty old power supply, and a serial number mystery”